The 6.3 release includes new features as well as improvements to existing functionalities.
In the Planmeca Romexis® 6.3 software version, we have implemented several improvements to the overall security of the software. Additionally, the security of Romexis 6.3 has been reviewed by a 3rd party cybersecurity services company,
Nixu Corporation. According to the review, the overall security status of Romexis is good (on a scale of good, moderate or inadequate). A summary of the review report is available upon request.
The main enhancements include:
Please note that the previous version of Romexis to undergo similar review was Romexis 5.1. We highly recommend that our customers update their Romexis software to the latest available version.
In Romexis 6.3, it is possible to open two studies and compare them side by side.
There is also a new toolbar for the image rotation and crop tools. The options in the toolbar include free rotate, rotate right 90, rotate left 90, crop, mirror horizontally and mirror vertically. Mirrored and rotated images are marked with the ! mark in the header.
Romexis 6.3 features new options for applying selections to all images imported at the same time.
In the tooth chart, the number of images is now shown for each tooth number.
The speed/responsiveness of 2D enhancements both with sliders and over the image adjustments has been greatly improved.
Importing 3rd party DICOM images has been improved with a more robust detection and processing of the greyscale values. The brightness and contrast adjustment range of Romexis has also been increased to better cater to 3rd party DICOM images.
There is a new option in the Implant alignment feature for aligning implants along the general abutment extension instead of the implant body.
Instead of a separate dialog, implant guide design now opens in full screen work mode with an integrated object browser.
In Romexis 6.3, the implant guide resolution can be chosen between normal, high and very high. The previous version only included normal resolution. Higher resolution enables creating a tighter implant guide. Please note, however, that the higher resolution increases the calculation time which also depends on the used computer.
The new release makes it possible to create guides with unconnected parts. The user can connect parts using the Create bar tool. This is a useful feature, for example, for fixation pins when the sleeve goes too far away from the scan.
The Remove Material tool has also been enhanced with new options. The user can now select either a round or a square shape tool. In addition, the depth of the removal tool can be limited, which enables, for example, holes on one side of a crown only.
The new software version entails better text rendering with the customisable text height.
It is now possible to set the guide transparency slider in relation to the IO scan and the guide.
Updates can now be downloaded directly on the Implant library webpage at Planmeca.com.